Two customers have reported that they experience occasional random reboots of their D180 or D100. It is only known to occur with specific configurations of Dynon products, namely D180 + D100 (or D10A) + AP74 w/servos, with all instruments networked together. Dynon has reproduced the problem and is implementing a fix. The temporary work-around is to remove one of the instruments from the DSAB network. A fix will be in Version 5.3 software which is scheduled to be out late February. The Service bulletin is located here.
If you have this configuration of instruments, and have experienced the rebooting, please let us know. Email us at email@example.com.